Notes on CTD/O2 Data Acquisition and Processing Using Sea-Bird Hardware and Software (as available).
Average rating votes
MetadataShow full item record
The predecessors of today’s CTD (conductivity-temperature-depth) instruments were introduced in the 1940s, with modern accurate instruments developed in the 1970s (Emery and Thomson, 1998). Accuracy standards for these instruments remain at the WOCE targets: 0.002°C (ITS-90) for temperature, 0.002 g kg-1 (TEOS-10) for salinity, and 3 dbar for pressure. Here we discuss some aspects of acquisition, processing, and calibration of Sea-Bird Electronics Inc. (hereafter SBE) 9plus CTD data. Oceanic measurement and calibration techniques are subject to improvement. This outline should be viewed only as a supplement to the existing literature, manufacturer documentation, and manufacturer instructions......
Title of ReportThe GO-SHIP Repeat Hydrography Manual: A Collection of Expert Reports and Guidelines. Version 1.
Editor(s) of ReportHood, E.M.
Series;NrICPO Publication Series;134
CitationMcTaggart, K. E., Johnson, G.C., Delahoyde, M.C.F.H.and Swift, J.H. (2010) Notes on CTC/O2 Data Acquisition and Processing Using Sea-Bird Hardware and Software (As Available). In, The GO-SHIP Repeat Hydrography Manual: A Collection of Expert Reports and Guidelines. Version 1, (eds Hood, E.M., C.L. Sabine, and B.M. Sloyan). 10pp. (IOCCP Report Number 14; ICPO Publication Series Number 134).DOI: https://doi.org/10.25607/OBP-1348
The following license files are associated with this item: