Method for Salinity (Conductivity Ratio) Measurement.
Average rating votes
MetadataShow full item record
The method described here is a modification of th e technique described by Stalcup (1991). Salinities for tens of thousands of samples were measured by the method during the WHP revisit cruises of the R/V Mirai . The method is one of the best methods which provide excellent precision onboard. The precision of this method inferred from re peat runs on standard seawater (SSW) of the International Association for the Physical Sciences of the Ocean (IAPSO) during the cruises was about 0.00001 for double conductivity ratio......
Title of ReportThe GO-SHIP Repeat Hydrography Manual: a collection of expert reports and guidelines, Version 1.
Editor(s) of ReportHood, E.M.
Series;NrICPO Publication Series;134
CitationKawano, Takeshi (2010) Method for Salinity (Conductivity Ratio) Measurement. In: The GO-SHIP Repeat Hydrography Manual: a collection of expert reports and guidelines, Version 1. (eds. Hood, E.M., C.L. Sabine, and B.M. Sloyan), 13pp. (IOCCP Report 14; ICPO Publication Series 134). DOI: https://doi.org/10.25607/OBP-1339